This help system is intended to provide you with all information that is necessary for setup, manual, and remote control of the Copper Mountain Technologies Vector Network Analyzer. Please use navigation tools on the left of the window to access the sections.
If you have any questions or comments, please contact us.
This manual covers the two-port models of the Analyzers controlled by the S2VNA software version 22.2.5. The Analyzer models are listed below:
This manual contains design, specifications, functional overview, and detailed operation procedures for the Analyzer, to ensure effective and safe use of its technical capabilities.
Maintenance and operation of the Analyzer should be performed by qualified engineers with basic experience in operating of microwave circuits.
Glossary — The abbreviations which are used in this document.