The top panel view of the Analyzers is represented in the figures below.
Part of the R60
The test port (type-N male 50 Ω) is intended for DUT connection. It is also used as a source of the stimulus signal and as a receiver of the response signal from the DUT.
The top panel is equipped with the READY/STANDBY LED indicator running in the following modes:
•Green blinking light is standby mode. In this mode the current consumption of the device from the USB port is minimum.
•Green glowing light is normal device operation.
External reference frequency is 10 MHz, input level is 2 dBm ± 2 dB, input impedance is 50 Ohm. Output reference signal level is 3 dBm ± 2 dB into 50 Ohm impedance. Connector type is SMA female.
External Trigger Signal Input allows the user to connect an external trigger source. Connector type is SMA female. 3.3v CMOS TTL compatible inputs magnitude have at least 1 μs pulse width. Input impedance is at least 10 kOhm. The External Trigger Signal Output port can be used to provide trigger to an external device. The port outputs various waveforms depending on the setting of the Output Trigger Function: before frequency setup pulse, before sampling pulse, after sampling pulse, ready for external trigger, end of sweep pulse, measurement sweep.